Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||22 March 2010|
|PDF File Size:||6.6 Mb|
|ePub File Size:||1.29 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the program string were corrected to make all data valid, this printout would result from the above program. Neither light on, indicates the front panel signal output is active.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
The following procedures compare the instrument operation to its specifications, listed in Table In order to use the Oven Option The lines should not be separated by more than ten divisions on the display.
X Drive Ramp Output. The A address is normally set at the factory to: Oscilloscope display should show one square wave per division, 5 divisions peak-to-peak vertical. Used for a signal common, as well as providing protection against electrical shock in case of a fault. The Microfiche package includes the latest Manual Changes supplement as well as pertinent Service Notes.
The following examples are given to illustrate the basic procedure for developing a program. The total quantity of a part is given the first time the part number appears.
By pressing ‘print’ button you will print only current page. The digital voltmeter reading should be —4, to —4.
HP A User Manual | 23 pages | Also for: B
Hewlett-Packard uses a two-section serial number consisting of a four-digit prefix and a five-digit suffix. This checks the output with no attenuation. Procedures for checking electrical performance are given in Section IV. Verify that ramp period variations do not exceed ten divisions. The data lines carry coded messages which represent addresses, program data, measurements, and status bytes. Returns A from remote to front panel control unless Local Lockout mwnual been programmed.
Amplitude Modulation Envelope Manua, Test —. Resolution of a dc offset entry with ac function is determined by the resolution of the ac amplitude. TB a 50 Hz. The voltmeter readings should be within the tolerances shown. Digital voltmeter reading should be – V to To begin a single sweep: It calls attention to an operating procedure, mxnual, condition or the like, which,if not correctly performed or adhered to, could result in damage to or destruction of part or all of the product.
AO – Past to Present.
Signal Output Front or Rear Panel: Component replacement and internal adjustments must be made by qualified maintenance personnel. If the program string contains 48 characters or more, the A will hold up the bus while it processes the 48 characters before accepting and storing the rest of the string. Contnillers – Devices that can specify the talker and listener s for an information transfer, The controller can be an active controller or a system controller.
HP 3325A Calibration Manual
Set the A to 5. This will not interfere manua the operation of the bus unless it was set to STBY while addressed to talk.
If amplitude modulation is ON when A functions other than sine wave are selected, the output may be gated, depending on the level of the modulation input. If reliability problems with U16 are encountered in these earlier instruments, refer to Service Note A Repeat steps d and e.
Standard 3325w High Voltage amplifier outputs are fused. Press the High Voltage Output key again to turn the option off. This test uses the control, ROM, and control clock circuits to perform the following checks: Total this column and enter in the Ex2 space. Approved for public release; distribution is unlimited. Address – The characters sent by a controller to specify which device will send information on the bus and which device s will receive information.
The remaining program can then constitute a second string. Then press the O or key momentarily to increase or decrease the value of that manuaal by 1.